Electrology is the practice of electrical hair removal to permanently remove human hair from the body. Electrolysis is the actual process of removing hair using electricity.
In electrolysis, a qualified professional called an electrologist slides a hair-thin, solid metal probe into each hair follicle without puncturing the skin (when inserted properly). Electricity is delivered to the follicle through the probe, which causes localized damage to the areas that generate hairs, either through the formation of caustic sodium hydroxide (the galvanic method), overheating (thermolysis), or both (the blend method).
Three methods are used in electrology: Galvanic, thermolysis, and blend. All three methods have their own merits, and one method is not better than another. The success depends on the skill of the electrologist, the type of hair being removed, the condition of the skin and the pain threshold of the client. All three methods, when properly performed, can be thorough at destroying the hair matrix cells, and leaving follicles incapable of regrowing hair.
This method is named after Luigi Galvani and uses a person's body as an electrolytic cell. Galvanic electrolysis was first reported in medical literature in 1875 by ophthalmologist Charles Michel as a method for removing ingrown eyelashes. A galvanic hair remover is essentially a positive ground power supply that delivers 0-3 milliamperes through the body. The follicular probe is the cathode of an electrolytic cell. Sodium hydroxide formed at the cathode by the process of chemical electrolysis kills the hair matrix cells. Modern galvanic hair removers automatically adjust the voltage to maintain constant current.
Another method is known as thermolysis, also called radio frequency (RF), shortwave or diathermy. Thermolysis was developed in the 1920s and first reported in medical literature by Henri Bordier. A thermolytic hair remover is essentially a radio transmitter, usually with an output of about 0-8 watts at a frequency of 13.56 MHz. RF energy emanates from the probe tip to tissue within about a millimeter. Thermolysis works by heating the hair matrix cells to about 48 to 50 °C (118 to 122 °F), causing electrocoagulation.
The practitioner selects a metal probe that slides easily into the hair follicle, usually the same diameter as the hair shaft or smaller. The probe is typically 50 to 150 µm (0.002 to 0.006 inches) for all three modalities. Care is needed to insert the probe at the same angle as the hair is growing out of the skin. The probe is inserted to the depth of the hair matrix, the site where hair is formed. The power and duration of the electricity are started at the lowest setting, then gradually increased until the hair comes out as easily as possible. If the patient experiences significant discomfort, the settings can be lowered.
Most practitioners will advise that complete removal of male pattern facial hair takes between 1 and 4 years, with an average treatment length of 2 years in case of one session per week, one hour per session. Removal of body hair works considerably faster.
Status of profession
In the United States, electrolysis is regulated in many states, requiring training and licensing.
Electrolysis as a profession faced new competition in the 1990s after laser hair removal was developed and promoted as a quicker and easier way to remove hair. However, the U.S. Food and Drug Administration (FDA) declared laser can only claim to reduce hair growth, not permanently remove it.
- Hair removal
- Intense pulsed light
- Michel CE. Trichiasis and distichiasis; with an improved method for radical treatment. St. Louis Clinical Record, 1875 Oct; 2:145-148
- Bordier H. Nouveau traitment de l'hypertrichose par la diathermie. Vie Med., 1924, 5:561
- Hinkel AR, Lind RW (1968). Electrolysis, Thermolysis and the Blend: the principles and practice of permanent hair removal. Los Angeles, CA: Arroway Publishers, ISBN 0-9600284-1-2
- "Laser Facts". U.S. Food and Drug Administration. Archived from the original on 25 July 2018. Retrieved 26 September 2018.
- American Electrology Association
- British Institute and Association of Electrolysis
- The Society for Clinical & Medical Hair Removal, Inc.